Omni BR-Cryo Cooling Unit
Cooling system to protect heat sensitive samples during homogenization.
Bead Ruptor 24 Cryo Cooling Unit Overview
- Compatible with the Bead Ruptor 24 Elite
- Adjustable cooling rate
- Compatible with liquid nitrogen or dry ice with ethanol
- BR-Cryo maintain sample temperatures from 4°C to ambient
- Prevent protein degradation
- Ensure RNA stability
- Prevent enzyme activation
BR Cryo Installation and Operation with Dry Ice.
BR Cryo Installation and Operation with Liquid Nitrogen.
|Cryo Chamber Volume:||1 L|
|Air Source Requirements:||Compressed medical dry air (water < 5ppm)|
|Air Source Pressure:||120 PSI|
|Cooling Source:||Liquid Nitrogen or dry ice in ethanol (liquid nitrogen is preferred)|
|Liquid Nitrogen Consumption:||1 L per 5 min of homogenization time|
|Dry Ice Consumption:||1 L of 0.5 L ethanol + 0.5 L dry ice per 90 minutes of homogenization time|
|Tube Compatibility:||With 1.5 mL, 2 mL and 7 mL tube carriages only. Not compatible with tube sizes >7 mL|
|Dimensions:||29.2 (high) x 34.3 (deep at base)/20.3 (deep at top) x 22.9 (wide) cm|
The BR-Cryo Cooling Unit is compatible with the Bead Ruptor 24 Elite and is designed to prevent the increase of sample temperature during the homogenization process. Friction generated through the bead milling process generates heat that is imparted to the sample. In order to maintain a constant processing temperature during homogenization, the BR-Cryo passes chilled air (-100°C to -50°C) into the sample processing chamber and across the sample tubes to dissipate heat. Samples can be maintained at a temperature ranging from 4°C to ambient thus protecting heat sensitive samples such as RNA and proteins. The BR-Cryo can be used to maintain a constant temperature when extracting functional enzymes or volatile small molecules.
Intra-tube temperature monitoring during bead mill homogenization.
2 ml Omni reinforced tubes containing five 2.8 mm ceramic beads (Cat # 19-628) were filled with 1 ml H2O and processed at 7 m/s on the Bead Ruptor 24 for increasing time periods. Homogenization was performed with and without the use of the BR-Cryo cooling unit using liquid nitrogen as the cryo source. Intra-tube temperatures were monitored at each time point. (in the last picture)
|Items||Ref. No.||Package||Price ex. VAT||VAT|
|Omni BR-Cryo Cooling Unit||19-8005||1 pc||€ 6 427.10||21 %||Buy|
(published 13.03.2019, size 732.7 kB)